Lipin/Dietz Associates, Inc.
OSHA 300 Log If you need to download an excel copy of the OSHA 300 log (in PDF Format) check this section.
Audio Booth Troubleshooting Tips
Audiometric booth troubleshooting If no sound is present through either earphone when your audiometer is connected to an audiometric test booth, check this section for some troubleshooting tips.
Audiometric booth unpacking instructions If you've just received a new AR-901 Sound room and want to know how to unpack it from the crate, check this section before you proceed.
Daily Biological Audiometer Calibration Form- Lipin/Dietz Occupational Health Format Click on this hyperlink to view and print copies of a sample Daily Biological Audiometer Calibration Form. This form was designed for use with most industrial occupational health programs.
Daily Biological Audiometer Calibration Form DOD Format Click on this hyperlink to view and print copies of a sample Daily Biological Audiometer Calibration Form in DOD format. This form was originally developed by the military.
Oscar Resources If your audiometer will not calibrate, check this section for hints on how to troubleshoot your audiometer connections and Oscar. You can also read about the theory behind an Oscar and how an Oscar works.
Tremetrics RA500 Audiometer
RA500 Data Transfer Protocol If you are looking for the data stream protocol to capture data from an RA500 audiometer, check this section for installation and operating instructions. This file is in adobe pdf file format.
Renaissance I Spirometer
Renaissance I Spirometer operating instructions If you need help using a Renaissance PB100 spirometer, check this section for some operating
Renaissance I Spirometer Printer problems If your HP Printer is not working with your Renaissance PB100 spirometer, check this section for some operating tips.
Renaissance I Spirometer -Deleting Records From Memory Patient reports that are saved on the memory card can be deleted as a group or one at a time. Check this section if you need access to specific instructions.
Renaissance I Spirometer operating instructions If you need help using a Renaissance PB100 spirometer, check this section for some operating tips
Renaissance II HP 5650 Printer operating Instructions If you need help using the HP 5650 Printer, this link is a PDF users manual for the printer.
Renaissance II Spirometer operating instructions If Click here if you need access to the complete Renaissance II users manual.
Titmus Vision Tester
Has your Titmus Vision Tester been serviced within the past year? If your Titmus Vision Tester has not been serviced in the past year, check out this section for some enlightening information.
3000 System Audiometer